External Remote-Controlled Frequency Down-Converter
Certificate of Federal Agency for Technical Regulation and Metrology – Approval of Measuring Equipment Type RU.E.35.018.A No. 48021 as of 11.09.2012. Entered into State Register of Measuring Equipment, Reg. No 51121-12.
Certificate of measurement procedure qualification No. 206/000265/2011 as 01.02.2011. It is also registered in the Federal Metrology and Standardization Foundation with No. ФР.1.38.2011.10001.
Certificate of measurement procedure qualification No. 002/2013 as 11.10.2013. It is also registered in the Federal Metrology and Standardization Foundation with No. ФР.1.35.2014.17680.
- Conversion of UHF and SHF signals to an intermediate frequency.
ARC-KNV4 Frequency Down-Converter
- Remote converter module
|Operating frequency range:|
|with internal antenna||3 – 18 GHz|
|with external antenna||1 – 18 GHz|
|Output signal frequency (IF)||70 MHz|
|Bandpass at -3 dB level||24 MHz, min|
|Relative error of sine signal frequency measurement within operating frequency range:|
|When using internal reference generator||±5x10-7|
|When using external reference generator||Depends on external reference generator parameters|
|Error of sine signal level measurement at external antenna inputs||±3 dB, max.|
|IP3 at SNR equal to 1 within 3 kHz bandwidth||68 dB, max.|
|Self-noise average level within 3 kHz bandwidth||-100 dB, max.|
|Standing Wave Ratio (SWR) at external antenna input with input resistance 50 Ohm||3 max.|
|Field sensitivity of buit-in antenna at SNR equal to 1 within 3 kHz bandwidth||max. 50 µV/m|
|Spurious rejection when using internal antenna||45 dB, min|
|Aperture angle of internal antenna at minus 3 dB level||100° max.|
|Operating temperature range||From 0°C to +50°C|
|Dimensions without handle||240×220×65 mm|
|Weight with handle||max. 1.8 kg|